Percutaneous device closure of atrial septal defect (ASD) is now considered the choice of treatment. Numerous devices with advantages/disadvantages are currently available and under development. Cardia Ultrasept II ASD occluder has a nitinol frame covered with polyvinyl alcohol (PVA) membrane. Here, a well-documented case of early malfunction of PVA membrane detected in the first week of implantation in a 4-year-old male patient, who underwent an uneventful device closure with 20 mm Cardia Ultrasept II ASD Occluder, is presented. One week after implantation left-to-right shunt through the device was detected and the explantation of device revealed PVA membrane with multiple perforations. © 2015 Wiley Periodicals, Inc.