Biomechanical study using human cadaver spines.Objective.
To assess the stabilizing effect of a supplemental anterior tension band (ATB, Synthes) plate on L5–S1 anterior lumbar interbody fusion (ALIF) using a femoral ring allograft (FRA) under physiologic compressive preloads, and to compare the results with the stability achieved using FRA with supplemental transpedicular instrumentation.Summary of Background Data.
Posterior instrumentation can improve the stability of ALIF cages. Anterior plates have been proposed as an alternative to avoid the additional posterior approach.Methods.
Eight human specimens (L3 to sacrum) were tested in the following sequence: (i) intact, (ii) after anterior insertion of an FRA at L5–S1, (iii) after instrumentation with the ATB plate, and (iv) after removal of the plate and adding transpedicular instrumentation at the same level. Specimens were tested in flexion-extension, lateral bending, and axial rotation. Flexion-extension was tested under 0 N, 400 N, and 800 N compressive follower preload to simulate physiologic compressive preloads on the lumbar spine.Results.
Stand-alone FRAs significantly decreased the range of motion (ROM) in all tested directions (P < 0.05); however, the resultant ROM was large in flexion-extension ranging between 6.1 ± 3.1° and 5.1 ± 2.2° under 0 N to 800 N preloads. The ATB plate resulted in a significant additional decrease in flexion-extension ROM under 400 N and 800 N preloads (P < 0.05). The flexion-extension ROM with the ATB plate was 4.1 ± 2.3 under 0 N preload and ranged from 3.1 ± 1.8 to 2.4 ± 1.3 under 400 N to 800 N preloads. The plate did not significantly decrease lateral bending or axial rotation ROM compared with stand-alone FRA (P > 0.05), but the resultant ROM was 2.7 ±1.9° and 0.9 ± 0.6°, respectively. Compared with the ATB plate, the transpedicular instrumentation resulted in significantly less ROM in flexion-extension and lateral bending (P < 0.05), but not in axial rotation (P > 0.05).Conclusion.
The ATB plate can significantly increase the stability of the anterior FRA at L5–S1 level. Although supplemental transpedicular instrumentation results in a more stable biomechanical environment, the resultant ROM with the addition of a plate is small, especially under physiologic preload, suggesting that the plate can sufficiently resist motion. Therefore, clinical assessment of the ATB plate as an alternative to transpedicular instrumentation to enhance ALIF cage stability is considered reasonable.