|| Checking for direct PDF access through Ovid
To compare fixation stability and fixation loss between the Humphrey Field Analyzer (HVF, static fixation target) and the RareBit computer-based perimeter (RBP, kinetic fixation target) during visual field testing.Fourteen healthy volunteer subjects wore an ASL Mobile Gaze Tracker as they completed HVF 10-2 and RareBit central field tests in a random order. Fixation stability, defined as the average distance from the fixation target to the subject’s gaze location, was calculated using data from the processed video capture. Fixation loss, defined as eye closure or a deviation of >20 degrees from the fixation target, was also measured. All subjects were surveyed regarding test preference.Use of the RBP kinetic target was associated with 18% improved fixation stability compared with the HVF static target (P=0.02). Nine of 14 study subjects demonstrated better fixation with RBP compared with HVF. Subjects demonstrated decreased fixation loss during RBP (0.9 s) compared with HVF (10.0 s) (P=0.002). Eighty-six percent of study subjects preferred RBP over HVF.Use of the RBP kinetic fixation target is associated with consistent fixation stability and decreased fixation loss compared with the HVF static target. This improvement in fixation stability may result from decreased perception interference (Ganzfeld, Troxler, and binocular rivalry effects), and may help account for the greater comfort reported with RBP compared with HVF.