These studies investigated the accuracy of the new Contour®Next ONE blood glucose monitoring system (BGMS) that is designed to sync with the Contour™ Diabetes app on a smartphone or tablet.Methods:
A laboratory study tested fingertip capillary blood samples from 100 subjects in duplicate using 3 test strip lots, based on ISO 15197:2013 Section 6.3 analytical accuracy standards. A clinical study assessed accuracy per ISO 15197:2013 Section 8 criteria. Subjects with (n = 333) or without (n = 43) diabetes and who had not used the BGMS previously were enrolled. Each subject performed a self-test using the BGMS, which was repeated by a site staff member. Alternate site tests and venipunctures were also performed for analysis. A questionnaire was provided to assess user feedback on ease of use.Results:
In the laboratory study, 100% (600/600) of combined results for all 3 test strip lots met ISO 15197:2013 Section 6.3 accuracy criteria. In the clinical study, among subjects with diabetes, 99.4% (327/329) of subject self-test results, 99.7% (331/332) of results obtained by study staff, 97.2% (309/318) of subject palm results, and 100% (330/330) of venous results met ISO 15197:2013 Section 8 accuracy criteria. Moreover, 97.6% (321/329) of subject self-test results were within ±10 mg/dl (±0.6 mmol/L) or ±10% of the YSI reference result. Questionnaire results indicated that most subjects considered the system easy to use.Conclusions:
The BGMS exceeded ISO 15197:2013 accuracy criteria in the laboratory and in a clinical setting.