To compare biometric changes measured with contact and noncontact methods after mitomycin-C–augmented trabeculectomy.Methods
In this prospective study, 31 eyes from 31 glaucoma patients scheduled for primary trabeculectomy were enrolled. Biometric parameters including axial length (AL), anterior chamber depth (ACD), and lens thickness (LT) were measured using contact ultrasound biometry (UD-6000 Ultrasonic A/B scanner biometer; Tomey Corporation, Nagoya, Japan) and a noncontact optical biometry device (Lenstar; Haag-Streit AG, Koeniz, Switzerland). Measurements were taken the day before trabeculectomy and then compared with measurements obtained 3 and 6 months after surgery.Results
The AL and ACD were significantly decreased at 3 and 6 months compared with baseline values taken with each biometry method. There was a significant increase in LT measured by the Lenstar device at the 3- and 6-month follow-up. At both the 3- and 6-month follow-up, the mean AL measurement reduction with the Lenstar device was significantly lower than that of the A-scan ultrasound measurements. The mean ACD changes between the two devices were not significantly different.Conclusions
There is a small but significant decrease in the AL and ACD after trabeculectomy as measured with both the contact and noncontact methods. The amount of AL reduction measured is significantly smaller using the noncontact method, making it the preferable method for intraocular lens power calculation for patients who need cataract surgery combined with or after trabeculectomy. The LT measured by the Lenstar device increased significantly after the operation, which can be an early sign of the progression of cataractous changes after trabeculectomy.