Practitioners fitting contact lenses for myopia control frequently question whether a myopic child can achieve good vision with a high-add multifocal. We demonstrate that visual acuity is not different than spectacles with a commercially available, center-distance soft multifocal contact lens (MFCL) (Biofinity Multifocal “D”; +2.50 D add).PURPOSE
To determine the spherical over-refraction (SOR) necessary to obtain best-corrected visual acuity (BCVA) when fitting myopic children with a center-distance soft MFCL.METHODS
Children (n = 294) aged 7 to 11 years with myopia (spherical component) of −0.75 to −5.00 diopters (D) (inclusive) and 1.00 D cylinder or less (corneal plane) were fitted bilaterally with +2.50 D add Biofinity “D” MFCLs. The initial MFCL power was the spherical equivalent of a standardized subjective refraction, rounded to the nearest 0.25 D step (corneal plane). An SOR was performed monocularly (each eye) to achieve BCVA. Binocular, high-contrast logMAR acuity was measured with manifest spectacle correction and MFCLs with over-refraction. Photopic pupil size was measured with a pupilometer.RESULTS
The mean (±SD) age was 10.3 ± 1.2 years, and the mean (±SD) SOR needed to achieve BCVA was OD: −0.61 ± 0.24 D/OS: −0.58 ± 0.27 D. There was no difference in binocular high-contrast visual acuity (logMAR) between spectacles (−0.01 ± 0.06) and best-corrected MFCLs (-0.01 ± 0.07) (P = .59). The mean (±SD) photopic pupil size (5.4 ± 0.7 mm) was not correlated with best MFCL correction or the over-refraction magnitude (both P ≥ .09).CONCLUSIONS
Children achieved BCVA with +2.50 D add MFCLs that was not different than with spectacles. Children typically required an over-refraction of −0.50 to −0.75 D to achieve BCVA. With a careful over-refraction, these +2.50 D add MFCLs provide good distance acuity, making them viable candidates for myopia control.